Laser hair removal handle repair for Alma and Syneron and many
1. Any brand,such as: Alma,Syneron and all Chinese brand,we can do
repair diode or handle
2. Diode Laser Bar from Germany
3. Repair more than 500 pcs hand piece each year
4. Our own cleaning room for mounting and assembling with best
Oriental-laser offer repair service for different brand,such as:
Alma,Syneron,Lumenis etc. And also all Chinese brand diode stack
and handle,such as: GT stack,FL diode,ADSS handle,Toplaser handle
etc.laser(Beijing)Co.,Ltd has a facility of 5,000 square feet of class
10K clean room & class 100 local clean zone (according to
Federal Standards 209E). oriental-laser has full production lines
of high power laser diode. We have professional technic team for
dealing with all technic issues.
We offer service:
1. Professional mounting and assemble serice
3. Rapid after-sales services.
4. OEM diode or hair removal handle for different machine demand.
Repair process as:
First, after we received the old broken handle piece from customer,
we check the appearance of handle piece to confirm whether all the
parts are complete and in good condition.
Second, we disassemble the handle piece, and test the old laser
stack. For the micro channel laser stack, Such as Alma laser,
Lightsheer, Syneron. We will test each laser plate one by one to
confirm whether there are any one bars still can work. If there
have some good bars, the repairing charge will lower. For macro
laser stack like most product made in China, we will test whole
laser stack together.
The third step
We choose best quality laser stack to replace the old broken one.
Our bars are import from Germany diode bar manufacture. We mounted
bars and assemble them to laser stack in our lab. We strictly
manage mounting, assembling and testing step during laser stack
producing. Every stack will be fully tested and had 8 hours burning
The Fourth step:
Our experienced technician will reassemble the handle piece. Which
a. Install the new laser stack into the handle piece.
b. Clean lens and check the TEC and other parts inside the handle
c. Test the power output of whole handle piece.
d. Clean the cover shell of handle piece and package it.
We do all above in our cleaning room: